Question about Measuring Tools & Sensors

# Where to buy a mitutoyo #513-518T dial indicator replacement cyrstal

Posted by on

• Level 3:

An expert who has achieved level 3 by getting 1000 points

All-Star:

An expert that got 10 achievements.

MVP:

An expert that got 5 achievements.

President:

An expert whose answer got voted for 500 times.

• Master

Your nearest TOOLKING shop will haveit for sure

Posted on Aug 01, 2009

Hi,
a 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of.(from cars to computers, handyman, and even drones)
Goodluck!

Posted on Jan 02, 2017

×

my-video-file.mp4

×

## Related Questions:

### I have a mitutoyo dial depth gauge. How do I use and read this?

A dial depth gage should have a set of depth stems. they are in 1" increments. to change the stem, first using a scale, measure the depth of the hole you are checking. Find the stem that falls in that range. remove the cap on the top of the unit, pull the existing stem out and replace it with the stem you need to reach the bottom of the hole. replace the screw-on cap. Set the cross bar of the body on the face of the part. Gently turn the barrel until the stem touches the bottom. the barrel has a 1" range. The numbers represent .1". The lines between the numbers represent .025". One turn of the dial equals .025" Add the length of the stem, the number on the barrel, the number on the dial and the number diall. Example, If the hole measures between 3" and 4" i.e. (3-1/8")start by installing the 3 to 4" stem. Turn the barrel so it reads "0". place the cross bar of the body so it touches the face perpendicular to the hole. Gently turn the barrel of the unit until it stops at the bottom of the hole. Add 3" plus the number on the barrel (.1"), the number of graduations on the barrel, and the number on the dial (.023"). The reading is 3.123". If you know your decimal equivalents, you will know that the hole is .002" short of 3.125" (or 3-1/8" deep.

Feb 21, 2011 | Mitutoyo 0-6"X.001" DIAL TYPE MITUTOYO...

### How do you rezero a 505-637 mitutoyo dial caliper

Fully close the caliper. The pointer should be pointing about straight up or close to it. Losen the knurled screw which is located on the circumference near the bottom. Now grab the dial and turn until the zero lines up with the needle. Re-tighten the knurled screw.

Dec 03, 2010 | Mitutoyo 0 150mm X .02mm Dial Caliper

### Im looking for the instructions manual for Mitutoyo 543-271B

For any manual contact Mitutoyo nearest office , they are professional and will help you.

Feb 13, 2010 | Measuring Tools & Sensors

### How to reset zero position reading on a mitutoyo 505 629 dial caliper?

unless they have changed them there is asrew like a watch winder on the dial it self ,undo it slightly and turn the dial to the desired position then retighten the screw

Nov 08, 2009 | Mitutoyo 505 677: Test & Measurement...

### Adjusting the angle of the contact point on a Mitutoyo 513-402

By simply pushing it (the contact point), it may be stiff to start but should ease up over time. Remember if your doing precision work (0.001mm or seconds of a degree) changing this original position will affect your reading so refer to booklet within original packaging or visit mitutoyo website's (Japan, USA, UK) between them they have most instructions available it's just a question of finding it, Good Luck.

May 09, 2009 | Mitutoyo Dial Indicator-Graduations:0....

### How to adjust Mitutoyo 505-101 caliber

You don't. What you did to adjust zero is what you do to adjust zero. Do not "go inside to adjust the dial". If the dial reads zero when the jaws are shut there is no problem.

Mar 18, 2009 | Measuring Tools & Sensors

## Open Questions:

#### Related Topics:

85 people viewed this question

Level 3 Expert

Level 3 Expert