Question about Drive Health & Beauty
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
SOURCE: mabis mist II not making mist
have you checked the wall plug to which you have tried to plug it in or maybe the cord to your nebulizer has a short in it somewhere and last maybe you need a newer model. Good Luck!
Posted on Jan 24, 2009
You'll read a lot about making sure that the filter is clean and that the medicine cup is cleaned and that hard water can effect the operation of the nebulizer. However, the problem is most likely in the pump. Even though the pump is running, it is not pushing enough air into the medicine cup. You may even feel air coming out of the tube. When unrestricted, the air will come out, but, once restricted by the medicine cup, the air may not be fast enough to nebulize the liquid. If you want to confirm this, take the tube and hold it to the end of a bike pump, if, while pumpung, you successfully see mist, then you know for sure.
Fixing the pump is not too difficult. Over time, the chamber where the air is compressed becomes dry and, just like a dry bike pump, the air escapes from the chamber due to poor seals. The pump is pretty simple. Make sure it is unplugged from power and remove the screws holding the housing together. Inside you will see the motor, a fan, a power switch and the intake and outtake tubes. Follow the outtake tube to the pump. The junction where the tubes meet the pump is the location of the air chamber. You may need to remove the screws that secure the tubes to the chamber. Once apart, take some grease and lightly coat all of the interior surfaces of the chamber. When finished, reassemble and you're good to go!
Posted on Apr 21, 2009
I, too, am looking for a manual for this product. I get a green light, but it does not appear to have any airflow coming from it.
Posted on Nov 05, 2010
Tips for a great answer:
Apr 02, 2013 | Medline Nebulizer Portable Compressor Sys...
Aug 06, 2012 | Omron Health & Beauty
Aug 06, 2012 | Omron Ne U22V Microair Vibrating Mesh...
Apr 27, 2011 | Omron Ne U22V Microair Vibrating Mesh...
Jan 25, 2011 | Sonic John Bunn Nano- Compact Ultrasonic...
Nov 12, 2010 | Respironics Inspiration Elite Compressor...
May 17, 2010 | Mabis 40 270 000 Mist II Ultrasonic...
23 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: