Question about The Health & Beauty
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Ok. Your not exhausting fast enough. Check your exhaust pipe to see if it tightly connected. Next check your roof to see if the exhaust is not blocked. If all those check out. Your heat exchanger is probably cracked or one of your burners is totally dirty...good luck. If your a novice, it would be best to contact a professional.
Posted on Apr 17, 2009
Innovage model hl 168J Blood Pressure Monitor. Problem: after the pressure build up and the meter starts dropping it goes to the Diastolic reading then records error. I notice the drop is very rapid. On other monitors I have used the drop is much slowe
Posted on Jul 25, 2010
Here are all the error code and their solutions
Error -- Possible causes Solutions
Display is blank
The batteries are dead. Replace both batteries with new
1.5 V R03/LR03/AAA batteries.
E 1 The cuff is not worn correctly.
Make sure cuff is worn properly and take a new
E 2 You moved while a measurement was being taken.
Sit still and take a new measurement.
E 3 The cuff is not attached correctly.
Fit the cuff correctly and take a new measurement.
E 4 An error has occurred in the measurement
Take a new measurement.
E 5 Cuff pressure above 300 mmHg
Check the fitting of the cuff (make cuff smaller) and take a new measurement.
E 6 Batteries are too weak
Replace both batteries with new 1.5 V R03/LR03/AAA batteries.
Hope this help you out. J
Posted on Jun 06, 2015
Tips for a great answer:
Sep 17, 2016 | Health & Beauty
Jun 07, 2015 | Advocate Redi-Code Duo Talking Glucose...
Mar 15, 2015 | Accu Check Aviva Plus Blood Glucose...
May 13, 2014 | Accu Chek ACCU-CHEK Aviva Meter
An E3 Error Code means the meter is sensing that blood has already been applied to
the test strip (or the strip has already been used).
What to do: Please make sure you see the flashing blood drop on the display before
adding the blood sample. Ensure that you are using a clean, new strip each time you
perform a test.
Jul 31, 2012 | Electrolux Washing Machines
Apr 20, 2011 | One Touch Health & Beauty
Jun 24, 2010 | Health & Beauty
37 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: