Question about Invacare Health & Beauty
a 6ya Expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repair professionals here in the US.
click here to Talk to an Expert (only for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Some chairs, like the M91 are a real bear to replace the batteries and connectors can get knocked loose in the process. First, using a voltmeter you need to see if the controller is getting power. On the DC setting, use voltmeter at the three-pin female connection at the joystick that is the charging port. The two pins that are opposite each other are the "hot" pins, and your meter should read about 25 volts, +- 2 volts. If its 0, there is a connection that isn't made from the batteries through to the controller, or the controller or joystick is shot. If you do have power to the joystick, then it's probable that the joystick is shot. A new one from Invacare is pricey, around $1,000.00 I have found a new site that can repair most makes. I have not used them, so cannot vouch for their work, but give them a call and see what you think.
If you don't have power, start at the terminals from the battery, and follow each wire back to the power control box, unplug and replug firmly as you work your way back. Check all fuses on the way. Then check all of the other connectors inside the chair, unplug, and reconnect again. Unplug and reconnect the wires leading into the power control box, and then restart chair. If no lights still, then the power control module is probably the culprit, but the address above fixes those too. The joystick and the power control module work together to make the chair operate safely. Good luck
Posted on Jul 08, 2009
The three light flash indicates a short circuit between the left motor(as you are sitting in the chair)and the battery. Check all the wiring from the control box to the motor and make sure it is not damaged in any way. If it is okay, you have most likely fried the left side motor. A new motor-gearbox assembly for this chair runs in the neighborhood of $1,400.00, or you can take the motor off and take it in for a full rebuild for about $250 or so. Unplug the joystick and remove the seat and cowl. Jack up the left side with both motors in gear. Remove the center nut cover and take off the nut holding on the wheel taking care to retain the key that slides into the groove on the axle. Disconnect the wiring plug to the motor and remove the six bolts holding the motor to the frame. Separate the gearbox from the motor and take the motor to a rewind shop for repairs. Reinstall and it should be good to go.
Posted on Jan 08, 2010
SOURCE: i am in need of
You can get the manual on the link below for your chair.
NOTE: The following procedures refer to the Pronto. R2 and the Pronto. R2-250 Series
Good Luck and please rate 4 thumbs up for the free answer.
Posted on Aug 13, 2010
Testimonial: "such a quick response is very appreciated an directions to the correct web site. thank you very much for a job well done."
SOURCE: Where is the reset button
Hello,you are my reason for being here,especially anyone with a wheelchair,i have a scooter myself & know what its like not to have your wheels.Please rate me as in thumbs up 1 to 4 for my answer.
Visit:WWW.ALLDATASHEET.COM Type in the right boxes-troubleshooting_2007 SHOPRIDER.pdf.It is 36 pages long 1 to 4 covers batteries,5 covers chargers,6 to 18 covers scooters & 19 to 36 covers wheelchairs & im sure you will be happy to see the part that you need to know about.Save it as a PDF File to your Adobe Reader + as i done print +save it to a polythene sleeve folder for future.Have a nice day i was so happy to have helped you,Colin.
Posted on Oct 08, 2010
Tips for a great answer:
May 07, 2016 | Invacare Health & Beauty
Dec 25, 2015 | Health & Beauty
Aug 23, 2014 | Invacare Pronto M51 Power Wheelchair
Jun 13, 2014 | Invacare Pronto M51 Power Wheelchair
Dec 15, 2013 | Invacare Pronto M51 Power Wheelchair
Jul 02, 2012 | Electric Mobility Invacare Pronto M41...
Feb 23, 2011 | Invacare Ivc Tracer Iv Wheelchair 22" ...
Jan 29, 2011 | Invacare Pronto M51 Power Wheelchair
May 02, 2010 | Invacare Supply Group Invacare Pronto M51...
23 people viewed this question
Usually answered in minutes!