Question about Health & Beauty
A 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
The service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
A monitor using Hg is only a valid monitor reading. Most electronic devices produce approximate results and are mostly good for taking several readings daily and obtaining a mean daily BP. Device calibration appears to be not accessible to the the user.Also, depending on age & condition of your dad, sequential reading must take place after elapse of enough time for circulation to be at steady state in the arm again, or if you immediately after a measurement make another, even with the same monitor you will obtain a different result.
Posted on Sep 07, 2009
E might represents insufficient inflation for your mom. For automatic inflation monitors usually there is a mechanism to keep inflating the cuff to higher pressures for people with higher BP. Check how is done. In some auto monitors,while cuff is inflated you keep pressing on START button until pressure reaches to some 15 to 20 mmHg higher than the pressure that auto usually stops and displays E.
Posted on Sep 08, 2009
Posted on May 14, 2010
I have digital blood pressure monitor Model 302 made in japan. It is not working. when start shoes error 3. Please advise
Posted on Dec 04, 2010
Possible explanations and work-arounds based on various blood pressure monitors..net is if it has worked religiously like mine had and this error comes up, and the simple fixes below do not work, there is something internally wrong with the unit..there does not seem to be any customer reset or obvious fixes..
Error Description Potential cause and remedy
«Err 3» No pressure in the cuff
An adequate pressure cannot be generated in the cuff. A leak
may have occurred. Replace the batteries if necessary. Repeat the
Err 0 No pulse is detected
Err 1 Inflation rate is too low
Err 2 Systolic reading unattainable
Err 3 Cuff pressure is too high
Err 4 Diastolic pressure is outside of operation range
Err 5 Cuff deflated too fast
Err 6 Cuff deflated too slow
Err 7 Systolic pressure is outside of operation range
Err 8 Overall pressure is outside of operation range
Err 9 Cuff is unable to inflate to the necessary pressure level
Error No. Possible cause(s)
Err 1 No pulse has been detected. Cuff may be positioned incorrectly.
Err 2 The arm was moved during the measurement.
Err 3 Inflation of the cuff took too long. The cuff is not positioned correctly.
Posted on Jan 19, 2011
Tips for a great answer:
Jan 10, 2015 | Microlife Health & Beauty
Aug 14, 2014 | Health & Beauty
Jul 08, 2014 | Health & Beauty
Jul 01, 2011 | Health & Beauty
May 29, 2011 | Health & Beauty
Mar 03, 2011 | Lifesource Digital Blood Pressure Monitor...
Sep 06, 2010 | Walgreens By Homedics Automatic Blood...
Feb 09, 2010 | Health & Beauty
Jun 28, 2009 | Microlife 3AC1-PC Blood Pressure Monitor
Mar 04, 2009 | Mabis Automatic Digital Blood Pressure...
66 people viewed this question
Usually answered in minutes!