Question about Contec Medical
A 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
The service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
SOURCE: What does the Power line
It means there is electronic noise coming from your electrical outlet. You can test to be sure by trying the unit unplugged from the wall. If it does not give the same error then it is definitely you power outlet.
Posted on Jun 03, 2009
Does it work in auto mode ? are you putting paper correctly means , there is a black mark on paper which is sensed by paper sensor .If you put paper up side down then paper sensor won't be able to sense any paper & gives you error for rifilling paper .If this doesn't help you then You need to replace battrey . Thanks
Posted on Jul 22, 2009
SOURCE: How does ECG machine find
When the at2 is in auto mode, first of all it check the leads are properly connected or not.Then it will acquire the ECG and store the same in its memory then only it starts printing. that's why it takes that much time to give a printout. When its starts printing you can remove the connections from the patient because it have already memorized the ECG.
If interpretation is in on position it will compare the acquired ecg with the standards in its memory (allready programed by the manufacturer). and will give a detailed report with ecg.
Posted on Feb 06, 2011
SOURCE: MY CONTEC ECG MACHINE CMS80
The power supply (That converts wall (mains) AC power to DC ) is not making "clean" dc power. The most likely problem is the large filter capacitors are " leaking" currect or dry. The small AC component (ripple) that is not filtered from the transformed dc can look like a heartbeat so there is some protection circuitry in these devices. It may be something else, but the large can filter capacitors are most likely. I always recommend that medical equipment be repaired by trained, insured factory technicians. Too many problems are likely to occur and you don't want to misdiagnose someone or electrocute someone because you are too cheap to get it repaired professionally. On the other hand, if you are in central Africa and using it for an elephant research project, get your soldering iron and replace the parts. -Mark
Posted on Sep 17, 2011
Tips for a great answer:
Jun 28, 2017 | Contec Medical
Jan 16, 2017 | Contec Medical
Dec 21, 2015 | Medical
Aug 18, 2012 | Contec Medical
Jan 14, 2012 | Contec Medical
Apr 12, 2017 | Ge Medical Systems Ge Marquette Mac 1200...
Apr 22, 2010 | Contec ECG Monitor & pulse oximeter
May 05, 2009 | Life Science Medical
Apr 15, 2009 | Medical
730 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: