Question about Schiller Welch Allyn Cardiovit At 2 Interpretive Ekg Machine
Hi the schiller at 2 plus ecg machine only prints a small amount when you press manual start and then the paper refill light keeps coming on although there paper in the machine
Does it work in auto mode ? are you putting paper correctly means , there is a black mark on paper which is sensed by paper sensor .If you put paper up side down then paper sensor won't be able to sense any paper & gives you error for rifilling paper .If this doesn't help you then You need to replace battrey . Thanks
Posted on Jul 22, 2009
Unproperly close lever may be ! try to press firmly, if still not work, clean print head with soft cotton, let it dry befor turn on .
Posted on May 27, 2013
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Aug 30, 2012 | Schiller Cardiovit At 2 Light Ecg Machine
Apr 27, 2012 | Schiller Welch Allyn Cardiovit At 2...
Apr 10, 2011 | Schiller Cardiovit At 2 Light Ecg Machine
Mar 05, 2011 | Schiller Welch Allyn Cardiovit At 2...
Feb 27, 2011 | Schiller Cardiovit At 1 Ecg Machine
Feb 15, 2011 | Schiller Cardiovit At 2 Light Ecg Machine
Oct 13, 2010 | Schiller Cardiovit At 2 Light Ecg Machine
Jun 19, 2009 | RTO Schiller CARDIOVIT AT-1...
May 13, 2017 | Schiller Welch Allyn Cardiovit At 2...
1,156 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: