Question about Health & Beauty
It worked fine for about a year. It stopped inflating about three months ago, and keeps showing Err message, even when I use brand new batteries.
Either the compressor unit seals are damaged or you have a hole in the inflatable section either way it is not an easy fix and as they have come down in price dramatically over the past year i would suggest buying a new model & chucking this one away
Posted on Nov 12, 2014
a 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of.(from cars to computers, handyman, and even drones)
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
SOURCE: I have a mabis 04-203-001
You must inflate far enough to stop blood flow, or the machine can not sense the systalic pressure. I suggest that you have your blood pressure checked. It looks like it my be HIGH!
Posted on Feb 06, 2009
If your HEM 704C fell hard, it is probably the pressure transducer that is faulty, broken. To my experience it is cheaper to buy a new one then to have it repaired by a technician
Posted on Jan 19, 2010
Testimonial: "Sound advice."
SOURCE: I keep getting an error code E83
How old is the cuff? If it has written on it "single use" and you've been using it a while you may have microleaks which means you need a new cuff. Also, check the hose for holes (usually found where the hose splits and connects to the cuff). Last, check the Dinamap to hose connector are the black orings still on the connecting posts?
Critikon/Ge utilizes the oscillitory method for determining NIBP. In other words, it pressurizes the cuff, stabilizes, decreases pressure slightly, stabilizes, etc. so a leak in the cuff or hose won't allow the pressure to stabilize resulting in an E83 error.
Posted on Dec 31, 2010
Tips for a great answer:
Jan 12, 2013 | HP Hewlett Packard Compaq 6720s Laptop...
Aug 04, 2012 | Omron Latest Model HEM-780 Automatic Blood...
Aug 10, 2011 | Asseenontvguys Wristech Blood Pressure...
May 21, 2011 | Computers & Internet
Jan 04, 2011 | Samsung Health & Beauty
Oct 08, 2009 | Sony Ericsson Walkman W705 Cell Phone
Oct 08, 2009 | Nikon DSLR D90 Digital Camera
430 people viewed this question
Usually answered in minutes!