Question about Salter 2001 electronic kitchen scale
My pelouze SP5 Digital Scale has out2 message. have removed the 2 screws, and released the interior bottom molded clamps, but the 2 pieces will not separate.
Posted by Anonymous on
a 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of.(from cars to computers, handyman, and even drones)
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
How to calibrate when "out2" is shown.
With the power off andthe scale on a stable surface, hold "g" and press "on". Some number will show up, probably a negative number. Press the "on" button again and the scale should display "Zero". Wait until it says "200.0". Gently place a 200 gram weight on the scale and wait a few seconds. If your successful it will say "Pass" and your scale will work again, yay!
Posted on Jun 21, 2010
How to open up the Salter 3007 Aquatronic Digital Scale:
1. Look for a hole (covered by a label) on the bottom of the device, not far from the L-R centre line, and about 1-3/4" from the front. Peel back the label and use a small cross-head screwdriver to loosen the screw that can be seen though the hole (needs about one turn, no need to remove it). Place the scale upright, and rotate the weighing pad slightly in a clockwise direction. This will disengage the tabs on the plastic pad which hold it to a metal base, so that it can be lifted off. 2. Remove the six screws recessed around the bottom of the scale. 3. Place the scale upright and gently "snap" apart the top and bottom shells (catches molded into the left and right ends). Be *** VERY CAREFUL *** when doing this as the display is held in place on a plastic bridge by the upper shell. When this is removed the display is loose except for a flexible plastic printed connector strip (which links it to the circuit board). It is *** all too easy for this strip to break away at one end or the other ***.
Posted on Sep 09, 2010
Tips for a great answer:
Apr 03, 2014 | Health & Beauty
Jan 19, 2011 | Salter 2001 electronic kitchen scale
Jul 27, 2010 | Tanita 1479V - Digital Pocket Scale /...
May 14, 2010 | Salter 2001 electronic kitchen scale
Jul 26, 2009 | DigiWeigh 100 X 0.01 Gram Digital Jewelry...
Jul 08, 2009 | RTO Pelouze Scale With Remote Display -...
Jun 15, 2009 | Balance Health & Beauty
Apr 10, 2009 | RTO Pelouze Scale With Remote Display -...
100 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: