Question about Invacare Health & Beauty
Posted by Anonymous on
a 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of.(from cars to computers, handyman, and even drones)
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
I believe I can, Please contact me at
firstname.lastname@example.org In subject put oxygen. will be looking to hear from you.
Posted on Jun 04, 2009
When these units are beeping it indicates they are basically not running. So you first need to ensure who ever is using it has access to o2 if they require. There isn't much you can do to fix these at home. You can first try going over the following. Not really that helpful but might help you sort it out. If you can't spot anything obvious ie unplugged no power etc then first contact anyone local such as the medical supply place. That would get you the fastest repair or help. If you don't have one of those to call then don't be afraid to call the 1-800 number below and ask the company.
The local company or even your doctor might be able to help you get bottled O2 while your unit is getting fixed.
If there is an audible alarm -- concentrator is not running: • Is unit plugged into a working outlet? • Check circuit breaker -- Right side of unit, behind a small door on the Companion 590 and 429a -- Back of the unit on the Companion 492 -- Between the on/off switch and flow meter
If there is an audible alarm -- concentrator is running: • Concentrator cannot be fixed by consumer • Call 314.362.0720 or toll-free 800.456.9366
There is a visual alarm on the Companion 590 with oxygen concentration indicator: • Green Light -- Normal -- Oxygen concentrations are correct • Yellow Light -- Below Normal -- Oxygen concentrations have fallen below an acceptable level • Red Light -- Not Operating -- Oxygen is no longer being produced; audible alarm will sound and concentrator will shut down
Posted on Apr 04, 2011
Tips for a great answer:
Jan 29, 2017 | The Health & Beauty
May 24, 2013 | Health & Beauty
Mar 28, 2013 | Air Purifiers
Apr 04, 2011 | Health & Beauty
Sep 01, 2010 | Fujitsu (G281065802)
May 11, 2010 | DeVilbiss 5 Liter Oxygen Concentrator with...
Mar 29, 2010 | DeVilbiss 5 Liter Oxygen Concentrator with...
May 25, 2009 | DeVilbiss 5 Liter Oxygen Concentrator with...
42 people viewed this question
Usually answered in minutes!