Question about Electric Jazzy 1115 Power Wheelchair
I have a invacare wheelchair with a mk5 joy stick that the lites wont blink i have new batterys in it
Some chairs, like the M91 are a real bear to replace the batteries and connectors can get knocked loose in the process. First, using a voltmeter you need to see if the controller is getting power. On the DC setting, use voltmeter at the three-pin female connection at the joystick that is the charging port. The two pins that are opposite each other are the "hot" pins, and your meter should read about 25 volts, +- 2 volts. If its 0, there is a connection that isn't made from the batteries through to the controller, or the controller or joystick is shot. If you do have power to the joystick, then it's probable that the joystick is shot. A new one from Invacare is pricey, around $1,000.00 I have found a new site that can repair most makes. I have not used them, so cannot vouch for their work, but give them a call and see what you think.
If you don't have power, start at the terminals from the battery, and follow each wire back to the power control box, unplug and replug firmly as you work your way back. Check all fuses on the way. Then check all of the other connectors inside the chair, unplug, and reconnect again. Unplug and reconnect the wires leading into the power control box, and then restart chair. If no lights still, then the power control module is probably the culprit, but the address above fixes those too. The joystick and the power control module work together to make the chair operate safely. Good luck
Posted on Jul 08, 2009
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Sep 09, 2015 | Invacare Pronto M51 Power Wheelchair
Jun 16, 2015 | Invacare Pronto M51 Power Wheelchair
The AMBER service indicator will light when an error or fault occurs. This is a yellow wrench shape. Refer to Diagnostic Codes on page 33 for a listing of the flash codes and what they indicate:
Mar 10, 2015 | Invacare Pronto M51 Power Wheelchair
Jan 23, 2015 | Invacare Pronto M51 Power Wheelchair
Aug 12, 2014 | Health & Beauty
Sep 23, 2013 | Invacare Pronto M51 Power Wheelchair
Mar 16, 2011 | Invacare Supply Group Invacare Pronto M91...
Oct 18, 2010 | Invacare Supply Group Invacare Pronto M91...
Jul 11, 2009 | Invacare Supply Group Health & Beauty
Apr 21, 2009 | Invacare Supply Group Invacare Pronto M51...
Jul 15, 2010 | Electric Jazzy 1115 Power Wheelchair
361 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: