Question about Invacare Pronto M51 Power Wheelchair
This is most likely a case that the controller/computer(the box that the joystick plugs into down inside the chair)is placing the chair into what is called a "limp-home" mode. It automatically senses something seriously wrong with the chair and starts cutting the power to the motors, such as overheating. If they are still getting hot, it will shut itself down completely, and you can restart after all cools down, but you have a serious problem on your hands. This usually happens when a chair is loaded near or past its maximum weight, and say you try to drive it up a long incline....motors will overheat and it will slow down, then shut down. If it is doing it on a flat surface without being heavily loaded, you have motor problems or controller problems.
Posted on May 08, 2014
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
SOURCE: oil / gel leak
If it is coming from your motors, it could be very bad. i.e. motor replacement. Sometimes on the side of those motors there is a black plastic cap about the size of a quarter or half dollar, those fall out sometimes and allow gearbox grease to come out. This should`nt be confused with your motor brush caps which are about the size of a pea and have a small **** in them for a screwdriver.DO NOT TAKE THESE OUT. If the large cap has come out just stick it back in. If you can`t find it you`ll ned to call a dealer to get the part replaced. If you can still find no prob,your probably going to have to go to a dealer.
Posted on Mar 26, 2009
The controller puts out fault codes though the LEDs. Count them and then look in the operators manual for likely causes. In my experience it is the electronic brakes that fail on these units and they are costly.
Posted on Oct 16, 2009
Hi. Your problem will be either the free-wheel lever is in the wrong position, or the motor has a stuck or worn carbon brush.
Good luck. Neil.
Posted on Jul 07, 2010
The 9 flash error code indicates a communications fault. The communication fault relates to the communication between the joystick and the control module. The joystick on the arm mount is connected to the control module in the base. When there is a break in the wiring of this wiring harness, also called a bus cable, a fault occurs. This trips the computer and the computer sends out a flash code to the joystick. Simply stated there is a broken or pinched wire between the joystick and controller. Look physically at the wire and check for cuts, breaks or pinches. The best short term repair is to repair the broken wire with solder and heat shrink or tape. You can also use **** splice connectors and tape. The best long term repair is to replace the harness with a part available from Invacare. The most definitive way to diagnose this problem is to read the fault log of the controller using a hand held programmer available to dealers and certified technicians. The number and type of fault codes logged in the controller is another indicator of which
component(s) has failed. Hope this helps.
Posted on Nov 20, 2010
If you have had swing away footrests on this chair you can replace them with model PHW93 and must ask to get them in red when you order them from a wheelchair store. If you dont know where a store is that sells Invacare products, call Invacare customer service at 800-333-6900.
Posted on Apr 07, 2011
Tips for a great answer:
Dec 25, 2015 | Health & Beauty
The AMBER service indicator will light when an error or fault occurs. This is a yellow wrench shape. Refer to Diagnostic Codes on page 33 for a listing of the flash codes and what they indicate:
Mar 10, 2015 | Invacare Pronto M51 Power Wheelchair
Aug 23, 2014 | Invacare Pronto M51 Power Wheelchair
Sep 25, 2012 | Invacare Pronto M51 Power Wheelchair
Sep 16, 2012 | Invacare Supply Group Invacare Pronto M51...
Oct 31, 2010 | Invacare Pronto M51 Power Wheelchair
Jul 11, 2010 | Invacare Pronto M51 Power Wheelchair
Jun 18, 2010 | Invacare Pronto M51 Power Wheelchair
Sep 10, 2009 | Invacare Pronto M51 Power Wheelchair
Apr 21, 2009 | Invacare Supply Group Invacare Pronto M51...
81 people viewed this question
Usually answered in minutes!