Question about Ge Medical Systems Ge Marquette Mac 1200 Interpretive Ecg
After we access the storage program (pressing the up/store/retrieve buttons simultaneously), high lighting print, send, delete or change doesn't do anything but kick us back to the main screen
Posted by Anonymous on
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
SOURCE: EKG Machine
3 lead machines are designed to monitor the cardiac rhythm cycle, in 3 of the 12 primary leads. An attempt to use this limited information as a true diagnostic tool would be incomplete and perhaps misleading.
A full 12-lead machine is diagnostic, and looks at the elecrical information from all sides ("walls", ie anterior, septal, inferior, lateral, posterior) of the heart and gives detailed information regarding ischemia and or infarction of the cardiac muscle, electrical conduction system function, electrical axis, chamber enlargement, drug effects, and more.
Hope this helps.
Posted on May 08, 2009
The heat on the print head can fade over time giving you a poor print. also the feed roller can wear and give poor pressure to the print head. thw way to diagnose it is to put a peice of electrical tape on the print roller and print an ekg. if it prints well in the area where the tape was, it is the print roller. if there is no change it is the print head.
email@example.com. www.overstockme.com. Jeff Watson
Posted on Jan 01, 2010
The battery is weak. Disconnect the AC, battery,& then replace the memory battery. Reconnect the power battery & AC. Next is a pain in the rear, reprogram the unit; date, ID, setup, ect.
Posted on May 14, 2010
Tips for a great answer:
Nov 09, 2011 | Ge Medical Systems Ge Marquette Mac 1200...
Aug 09, 2011 | Ge Medical Systems Ge Marquette Mac 1200...
Jul 11, 2011 | Ge Medical Systems Ge Marquette Mac 1200...
Jan 20, 2011 | Ge Medical Systems Ge Healthcare Mac 1200...
Jan 07, 2011 | Ge Medical Systems Ge Marquette Mac 1200...
Apr 19, 2010 | Ge Medical Systems Ge Marquette Mac 1200...
Dec 27, 2009 | Ge Medical Systems Ge Marquette Mac 1200...
May 28, 2009 | Ge Medical Systems Ge Healthcare Mac 1200...
Apr 22, 2009 | Office Equipment & Supplies
Apr 16, 2009 | Ge Medical Systems Ge Marquette Mac 1200...
Jan 29, 2015 | Ge Medical Systems Ge Marquette Mac 1200...
53 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: