Question about DigiWeigh 100 X 0.01 Gram Digital Jewelry Pocket Scale
F--H APPEARS WHEN TURNED ON AND NOTHING MORE WILL HAPPEN
That dosent work in the case of an f--h display at least in my case i think the f--h stands for f**king hopeless
Posted on Jul 18, 2010
I thought the scale was just completely broken, I was able to fix by doing the following.
1. turn unit off
2. hold down the on/off/tare AND the MODE button.
3. The unit will come up with a bunch of numbers on the screen. Release the buttons
4. Press MODE once, you will see "100" flash on the screen. if you don't press the button a second time.
5. place 100g of tare weights on the scale.
6. Press MODE button again. After this my scale said PASSED and shut off.
7. remove tare weights and turn device on. It should now register properly.
In my case. the scale was misregistering the 50g weights as 75. I shut it off. I pulled up ever so slightly on the scale plate on the hinge side of the plate are repeated steps 1-7.
I have no idea why they don't have a 3 part calibration on these.
It is working like a charm now!
Posted on Oct 03, 2009
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
Tips for a great answer:
Nov 08, 2016 | Dehumidifiers
Jun 08, 2014 | Universal MX-450 Remote Control
Jan 31, 2011 | Acer Aspire One Netbook 8.9" Notebook
Feb 07, 2010 | Apple 24" iMac Desktop
Jan 06, 2010 | Garmin nuvi 1300 Car GPS Receiver
Aug 26, 2008 | HTC Touch P3450 Smartphone
Aug 02, 2008 | Nokia 6111 Cellular Phone
Apr 14, 2007 | Philips GoGear SA261 MP3 Player
Mar 28, 2018 | DigiWeigh 100 X 0.01 Gram Digital Jewelry...
2,653 people viewed this question
Usually answered in minutes!