Question about Welch Allyn Spot Vital Signs Monitor with NIBP and Stand
BP dosent work when i power up, pulse ox works and temp works get error code CO2
Posted by Anonymous on
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
This link allows you to review error codes and gives suggested corrective action....
I hope this helps.
Posted on Jan 07, 2009
There is not enough current in the battery to start the pump. Even when plugged in these will not work properly if the battery is going bad. These are not technically user replaceable but with a screwdriver and a new battery you can take care of it. I can help more if you like. email@example.com
Posted on Aug 31, 2009
Tips for a great answer:
Oct 08, 2012 | Welch Allyn Atlas Monitor W/Ecg, Nellcor...
Feb 28, 2011 | Welch Allyn Spot Vital Signs Nibp/Nellcor...
Power the Spot Vital Signs
off, wait five seconds, and power on. If the error code
reappears then power the
Spot Vital Signs off and disconnect the battery for five minutes.
Reconnect the battery and
power on. Press the Blood Pressure Start/Stop button to reset
flashing patient alarm conditions.
Dec 29, 2010 | Welch Allyn Spot Vital Signs Monitor with...
Dec 21, 2010 | Welch Allyn Spot Vital Signs Monitor
Aug 11, 2010 | Welch Allyn Health & Beauty
Feb 10, 2010 | Welch Allyn Spot Vital Signs Monitor with...
Dec 10, 2009 | Welch Allyn Spot Vital Signs Monitor with...
Sep 30, 2009 | Welch Allyn Vital Signs Monitor with Nibp...
Nov 13, 2008 | Welch Allyn Spot Vital Signs Monitor with...
May 21, 2014 | Welch Allyn Spot Vital Signs Monitor with...
89 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: