Question about Medical
It's one of two things, either:
1) Your thermal printhead is dirty or going bad. Rub a pencil eraser over the printhead to clean it.
2) There's not enough pressure between the thermal printhead and the advancing assembly.
While printing, press the sliding door towards the unit, does the print get darker? If so, the mechanism might not be holding tight which indicates the clip may be damaged. If it stays light, clean the printhead with the eraser as mentioned earlier..
Posted on Mar 25, 2014
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Feb 17, 2016 | BURDICK Atria 6100 EKG
Oct 08, 2012 | Welch Allyn Atlas Monitor W/Ecg, Nellcor...
Dec 02, 2011 | Medical
Jun 15, 2011 | RTO Quinton Cardiology Atria 6100 Ecg...
Jan 24, 2011 | Medical
Dec 23, 2010 | RTO Quinton Cardiology Atria 6100 Ecg...
Sep 27, 2010 | RTO Quinton Cardiology Atria 6100 Ecg...
Mar 02, 2010 | RTO Burdick Quest Stress Test System w/...
Jun 20, 2017 | Ecg Medical
169 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: