Question about Topcon At-g2 Auto Level
Posted by Anonymous on
Fix the instrument to a solid tripod. Level the vial. Turn the instrument 180 degrees. If it is out of level. Adjust half of the bubble's error with the foot screws, and the other half with the 3 adjustment screws around the vial itself. Continue to go back and forth 180 degrees until the two sides match.
It will take some time to get the hang of. But is not hard to accomplish.
Posted on Nov 26, 2014
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Sep 13, 2016 | Toro Garden
Sep 10, 2014 | 1999 Pontiac Grand Am GT
Oct 28, 2013 | Sokkia Abney Level with Clinometer 804710
Apr 23, 2011 | Garden
Feb 16, 2010 | Stanley Hand Tools
Dec 21, 2009 | RTO CST/Berger 55 - SLVP32N 32X SAL...
Sep 05, 2009 | David White 44 8834 1 22x Level Transit...
Nov 25, 2008 | Starrett Measuring Tools & Sensors
Oct 21, 2008 | Stanley Measuring Tools & Sensors
Aug 02, 2008 | Nikon Optics
109 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: