Question about Drive Geo Power Wheelchair
Thanks for your question!
I'm sorry that you haven't received an answer yet.
The response rate on FixYa FREE questions is usually over 80%, but sometimes questions go un-answered when:
1. the question is unclear, incomplete, or poorly worded...
2. the question is mis-categorized - so the correct experts aren't viewing it...
3. the question requires an answer for something that cannot be explained online adequately...
4. specific training, tools, or equipment are required to avoid injury or death...
5. the Experts that are online simply don't have the resources to answer your question fully...
Please feel free to re-work your question if you feel one of these may be the reason.
Of course you are not required - but converting your question to a PAID one usually increases the reply rate as well.
Normal rates for FixYa are...
$ 9.99 answer for one question.
$12.99 Live Real-Time online chat or premium e-mail support.
$29.99 Live Real Time phone support
Sometimes FixYa will run specials for unlimited questions via monthly subscription or reduced rates for limited periods.
Consider the savings that can be had when an informed Expert can save you a $100 service call, or a $350 trip to the car dealership.
Your problem solved - guaranteed! Or your money back!!
If you don't want to pay - feel free to re-submit your question so it returns back to the top of the work queue if you still need help.
Thanks for using FixYa - a 4 THUMBS rating is appreciated for helping with your FREE question
Posted on Sep 05, 2010
A 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
The service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Sep 02, 2014 | Invacare Pronto M51 Power Wheelchair
May 27, 2014 | Electric Mobility Invacare Pronto M41...
Oct 24, 2013 | NRS Used Merit MP 3CR Power Wheelchair
Jul 11, 2012 | Drive Geo Power Wheelchair
Oct 29, 2011 | Ctm HS-2800 Power Wheelchair
May 29, 2011 | Electric Mobility Geo Wheelchair Drive...
Feb 26, 2011 | Electric Mobility Invacare Pronto M41...
The Drive Cirrus Plus power wheelchair is a folding frame chair with a VSI controller. The VSI controller (joystick) will indicate a fault code when it detects a problem with your power chair. It is manufactured by a company called Penny Giles Drive Technology. An LED flash code of 2 on the battery display indicates that the left motor is disconnected. There may be a brake fault also.
The left motor can be disconnected at the plug where the VSI harness plugs into the motor plug. The pins for the motor or VSI output may be pushed back in the connector or not making contact due to corrosion or dirt. Unplug this connector and inspect the pins and make sure they are making good contact and are clean. Spray contact cleaner on the contacts to clean or re-orient the pins to make better contact. The folding frame of the chair may have dislodged the connector. The wiring harness from the motor may be pinched or cut. The brake wires and pins may have become disconnected also. The brake lever may be in the disengaged freewheel position causing a fault, so put the brake lever in the correct position. Strictly speaking, this fault code is for the motor only, but always check the motor/ brake assembly since the wiring runs from the same plug.
A certified technician will have a hand held programmer to read the computer (VSI) fault log and determine the number of error codes for the various components.
Using a digital voltmeter you can check the motor at the plug. The two large wires on this 4 pin plug go to the motor; the two small wires go to the brake. On the resistance scale the motor should read between 0 and 1.6 ohms. The brake should read between 30 to 60 ohms.
There is a chance that the output of the VSI controller is bad. To test this swap the left and right motor output connectors from the VSI. If the error code now indicates right motor disconnected (flash code 4) you have a good VSI and bad left motor. Swap the leads back after you are done with this test.
Try blocking up the chair and see if the motor runs with no load on it or after spinning the drive wheel by hand. You can also inspect the motor brushes for burning and wear. If you pull the brushes, do this carefully and blow out the dust from the motor armature. This may allow the motor to work again for a short time.
You will probably need to hire a technician if, after checking for bad connections and the brake lever in the wrong position, you still have problems. I like to replace the brake/ motor/ gearbox as a complete assembly to avoid future problems. Labor for the diagnosis and reading the fault code log with a hand held programmer should be under 1 hour. A return visit to replace the motor brake gearbox is under 1 hour. Hope this helps.
Dec 25, 2010 | Drive Medical Cirrus Plus Folding Power...
Nov 24, 2010 | Ctm HS-2800 Power Wheelchair
May 18, 2009 | 1995 Geo Tracker
141 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: