Question about Clothing Accessories
Posted by Anonymous on
A 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
The service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
The zero capture is out of the 2percent tolerance so the cell is either faulty or requires zeroing. Open the unit and press the micros itch on the left of the board. The display flashes s. Press the zero key to change to flashing E. Press store to scroll to zero balance. Press semi auto tare. This captures zero without clearing the span calibration. The display will scroll around for a short period when complete press store.. Release the micro switch and scale will go thru warm up routine then display zero. Hope this helps, regs Darren.
Posted on Apr 15, 2012
I thinkn there is a reset button to press with no weight on the scale if I remember correctly. Might have to calibrate with the test weight too if the calibration is messed up (not likely though).
Posted on Jan 03, 2015
There is a similar problem on this site and solution.
My digiweigh scale reads HHH What does this mean and how do fix it
Posted on Feb 09, 2015
Tips for a great answer:
Feb 09, 2015 | Health & Beauty
Feb 06, 2015 | Health & Beauty
Jan 03, 2015 | Health & Beauty
Jan 02, 2015 | Health & Beauty
May 21, 2014 | RTO Pelouze Scale With Remote Display -...
May 17, 2014 | NATASHA COLLIS Gold nugget ring with mixed...
Apr 12, 2014 | DigiWeigh 100 X 0.01 Gram Digital Jewelry...
Oct 14, 2012 | Health & Beauty
290 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: