Question about Midmark Medical
Our midmark ekg machine will not consistantly record - the first 3 QRS complexes show then it goes flatline. is this a problem with the leads, or other probem?
Most likely you will have expired or cheap electrodes. Even if they stick to the patient there still may not be enough moisture in the electrode to conduct a signal. Either that or some of the lead wires may be damaged. All it takes is one intermittent lead to cause the problem you are asking about. The best thing to do is to start with he electrodes and then try the leads.
Posted on Jun 16, 2009
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Jan 30, 2014 | Ge Medical Systems Ge Marquette Mac 1200...
Aug 18, 2012 | Contec Medical
Mar 31, 2011 | Schiller Welch Allyn Cardiovit At 2...
Dec 06, 2010 | Schiller Cardiovit At 1 Ecg Machine
Dec 03, 2010 | Schiller Welch Allyn Cardiovit At 2...
Oct 13, 2010 | Ge Medical Systems Ge Marquette Mac 1200...
Sep 07, 2009 | Schiller Welch Allyn Cardiovit At 2...
Aug 11, 2009 | Medical
Aug 27, 2008 | Welch Allyn Basic Aed 20 with Non...
Aug 27, 2008 | Omron Portable Ecg Ekg Handheld Hcg 801...
230 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: