Question about Measuring Tools & Sensors
Terminal is dirty inside and unit thinks that leads are connected to mA terminals and knob is at Volts,
Use compressed air to clean terminals and if that does not help use cotton bud with little alcohol on it to clean terminals inside. It will fix the problem.
Posted on Sep 25, 2013
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Nov 03, 2014 | Motorcycles
Nov 22, 2013 | Velleman DVM850BL Digital Multimeter
Jan 24, 2013 | Fluke Networks Fluke 87-5 Digital...
Jan 08, 2012 | Cars & Trucks
Jan 08, 2012 | Measuring Tools & Sensors
Oct 28, 2010 | Sony Playstation 3 (PS3) Console
92 people viewed this question
Usually answered in minutes!