Question about Health & Beauty
Posted by Anonymous on
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
Either do a search for that particular scale brand & model, which should take you to the company, whom should have a manual on-line or go back under search and type in "users manuals". There should be several manual websites that come up. Just don't click on the websites that require money for their manuals, when you can get them for free at other sites.
Posted on Jan 22, 2009
Take a 5 pound bag of sugar, flour or what have you. Put it on the scale. If it is off, turn the scale so that you can check every possible place for an adjustment. Adjust and go for it!!! Good luck...Lee
Posted on Jan 17, 2010
The zero capture is out of the 2percent tolerance so the cell is either faulty or requires zeroing. Open the unit and press the micros itch on the left of the board. The display flashes s. Press the zero key to change to flashing E. Press store to scroll to zero balance. Press semi auto tare. This captures zero without clearing the span calibration. The display will scroll around for a short period when complete press store.. Release the micro switch and scale will go thru warm up routine then display zero. Hope this helps, regs Darren.
Posted on Apr 15, 2012
Tips for a great answer:
Apr 16, 2014 | Tanita Health & Beauty
Oct 11, 2013 | Homelite Garden
Oct 14, 2012 | Health & Beauty
Feb 04, 2012 | Health & Beauty
Mar 22, 2010 | Health & Beauty
Jan 16, 2010 | Jobar International Idea Works Wireless...
Jan 04, 2010 | Health-O-Meter Health o meter 402KLWH...
Dec 09, 2009 | Berkley Twist Lock Rod Rack
26 people viewed this question
Usually answered in minutes!