- If you need clarification, ask it in the comment box above.
- Better answers use proper spelling and grammar.
- Provide details, support with references or personal experience.
Tell us some more! Your answer needs to include more details to help people.You can't post answers that contain an email address.Please enter a valid email address.The email address entered is already associated to an account.Login to postPlease use English characters only.
Tip: The max point reward for answering a question is 15.
The best day chessboard type is the traditional type of weight, in contrast, electronic weight scale error degreehighestelectronic tems, affected by environments, bumps can make reading deviation.The use of electronic weight meter can more precisely? Electronic scales are generally placed in the hard flat floor use, not on the carpet or other soft object surface by using.Each weight to body weight in the same place, the same time, wear the same clothes. Hello, I just see a online stores http://www.tmart.com/Digital-Scales/ sell digital scales. Some advice, for reference purposes only and finally I wish you luck!
magnum 1000xr pocket scales turns on calibrates goes to different modes but wont show the weight when object placed on the scale,a lock appears in top left of window, is scales broke or do i have to make lock symble dissappear?
Here are the basic ERR messages and their meanings:
Error 0: Initialization
Error 1: Instability (scale is on an uneven surface ex: carpet)
Error 2: Unit may have been jarred or unit was overloaded (customer went over the maximum weight capacity)
Error 3: Contact error (Feet were not wet for Body fat reading, feet aren’t big enough to reach sensors)
Error 4: Body fat is out of range
Error 5: Water is out of range
I got an ERR2 message and now the scale doesn’t work. What should I do?
The following reset procedure is used when the scale shows an err2, err, 0.0, inaccurate weight, or some other unusual error.
Remove the battery from the scale.
Sit the scale on a hard surface floor.
Step up onto the scale, stand still for about 5 seconds and step off the scale.
Re-install your battery.
Place the scale back on a hard surface floor.
With one foot only, place just enough weight on the scale for the display shows zero’s or dashes. Once you see zeros or dashes remove your foot.
Wait for the scale to turn off again.
Step up onto the scale for a weight.
If the reset is successful, you should receive a weight right away. If you continue to receive one of the error messages above please call our Customer Service department or contact us via email. You can also try the reset procedure a 2nd time. Sometimes the reset procedure will work the second time.
The scales display EE as they are reading a negative weight.
With the scales switched OFF place 2 glass plates on the scales and press ON.
The scales should now display 0g.
If this doesn't work add more weight and try again.
There is also 4 screws on the underside of the scales which hold the weighing bar in place, if you loosen the 2 smaller screws this should relieve some of the stress so less weight is needed to zero the scales.