Question about Starrett T216XRL-1 - MICROMETER, DIGITAL
I have two 1" digital micrometers model 216-XRL-1.These have been opened past the 1" range, and now when they are closed they are off abbout .050. How can I repair these.ThanksBob
You may have damaged these beyound what you can repair. best thing woudl be to return to Starrett to see if they can be fixed. They will advise repair charge before repairing so you can decide. digital micrometers should not be opened beyount the range.
Starrett Tech Support
Posted on May 15, 2009
Micrometers must be calibrated at 4 spots around the barrel and at zero and full scale to be
truly accurate. You will have to send it to a calibration lab or back
to Starrett for a full calibration because they will check it with gauge blocks of known length and verify the spindle and anvil surface flatness.
If you are only concerned with the zero and not as much about the full accuracy of the tool, then you can reset the zero yourself. With the mic fully closed (spindle and anvil touching), turn the barrel with the funky, half circle shaped wrench that came with the mic in the box. There is a little hole in the barrel where the nub on the wrench will fit then you just turn it until it reads zero. Good luck.
Posted on Mar 07, 2009
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Feb 26, 2015 | Yamaha Motorcycles
Feb 26, 2015 | Mitutoyo 317-351 Digital...
Jan 08, 2015 | Fowler Measuring Tools & Sensors
Dec 26, 2014 | Yamaha Motorcycles
Jun 29, 2011 | Mitutoyo Depth Micrometer 0-6"
Apr 30, 2011 | Mitutoyo 193 211 Digital Micrometer
Mar 08, 2011 | Fowler Outside Micrometer Set 0 - 4"
You may check the calibration by using the following directions. First, adjust
your micrometer setting to 2. Squeeze the trigger against the handle. The
calibration line is a small, raised line embossed on the top of the spreader's
shut-off plate. With the micrometer set at 2 and the trigger squeezed against
the handle, the calibration line should be aligned with the back edge of the
To adjust the alignment, use a small wrench to turn the black calibration screw at the lower end of the control wire leading from the micrometer rate setting barrel. Adjust the screw until the calibration line and back edge of the hopper meet. If you have any difficulty checking the calibration, contact us and we will be happy to help walk you through the adjustment.
When you are not pulling on the trigger handle, the cable should be loose. When you pull on the handle, the cable becomes taut and the shutoff plate slides open.
If you need further assistance feel free to contact us @ 1-800-543-8873
Jan 24, 2011 | Scotts Company Scotts& Accugreeen 2000...
Jan 24, 2011 | Scotts Company Garden
Nov 08, 2009 | Scotts Company Scotts& Accugreeen 2000...
Aug 27, 2011 | Starrett T216XRL-1 - MICROMETER, DIGITAL
409 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: