Question about Motic AE30 Binocular Microscope
After setting up the microscope a while ago, it just worked really well until today. We didn't change anything on the microscope, but when you switch it on there is no lightbeam.
The light of the switch in illuminated and also the intensitybar lights up.
I already exchange the lightbulbe with a working microscope and still no light beam.
Can anyone tell me what the problem is?
Thans in advance.
a 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of.(from cars to computers, handyman, and even drones)
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
You can send the lens to Delta Optical Instruments, Inc.They can determine if your EF Plan 40x objective lens is repairable.The will give you an estimate prior to work or suggest a replacement if it cannot be repaired.email them at: email@example.com
Posted on Dec 25, 2010
Tips for a great answer:
May 16, 2011 | Computers & Internet
Mar 01, 2011 | Motic DS300 Monocular Microscope
Feb 03, 2011 | Motic DS300 Monocular Microscope
Jan 01, 2011 | National Optical DC2-155 Monocular...
Dec 01, 2010 | Boreal WW5790015 Microscope
Nov 16, 2010 | National Optical DC2-155 Monocular...
Feb 17, 2010 | Motic DMB5 Binocular Microscope
Dec 26, 2009 | Motic B1-220A Binocular Microscope
Sep 24, 2009 | National Optical DC3-163 Binocular...
38 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: