Question about Toshiba E-STUDIO 550 Copier
My Toshiba e-studio 550: service recommended for SPC, please provide the code on how to reset it, thank you
Check the control status of the surface potential sensor (05-242).
<When “4” is displayed: sensor error>
The sensor detection value is abnormal or the sensor output value is not changed even though
the main charger bias value is changed.
Is the connector of the surface potential sensor connected properly?
Is the main charger grid/wire attached poorly?
Is the main charger attached poorly?
Are leaks and such occurring?
Posted on Mar 06, 2009
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Jan 26, 2011 | Toshiba e-Studio163 Copier
Jul 02, 2010 | Toshiba E-STUDIO 550 Copier
Jan 19, 2010 | Toshiba Estudio 200L Copier
Jul 22, 2009 | Toshiba E-STUDIO 650 Copier
Mar 31, 2009 | Toshiba E-STUDIO 550 Copier
Mar 09, 2009 | Toshiba E-Studio 161 Copier
Jan 22, 2009 | Toshiba E-STUDIO 550 Copier
May 22, 2008 | Toshiba E-STUDIO 550 Copier
1,145 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: