Question about Science & Laboratory
I have a Carlsan single occular compound microscope. The course and fine adjustment knobs raise or lower the stage to focus. Lately once I have my slide in focus and I let go of the adjustment knob the stage level begins to slowly decend thus unfocusing the image. As this is happening I can see the course adjustment knob slowly spin. The ends of the fine adjustment knobs have heads that look like they could be adjusted with a flathead screwdriver but the ends of the course adjustment knobs have no such head. It seems like I need to get some more tension between the knobs and the track that they move on. I appreciate any help.
Try holding the Right knob so it does not move, whilst firmly moving the Left knob towards you (clockwise), this is the adjusting method used on Zeiss Jena 1960's High End microscopes, e.g Eduaval & Ergaval.
Posted on Mar 20, 2009
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Mar 19, 2016 | Nikon Labophot trinocular Microscope...
Oct 07, 2015 | Science & Laboratory
Dec 01, 2010 | National Optical 131 Monocular Microscope
Jul 08, 2010 | Swift M2251C Monocular Microscope
Feb 19, 2010 | Celestron 44320 Trinocular Microscope
Dec 09, 2009 | Motic DMBA 300 Binocular Microscope
Nov 19, 2009 | Boreal WW3111900 Microscope
789 people viewed this question
Usually answered in minutes!