Question about Conair Weight Watchers Glass Digital Body Fat Analysis Scale - 1 ea
My Conair Weight Watchers Model WW42 came up with the above error message. Figured batteries were dead. Replaced batteries, went to use scale and same error message came up.
Is my scale defective?
My scale had the same problem. I replaced the battery, but still got the same message. My wife noticed that one of the feet on the bottom of the scale was loose. The feet are twisted on to the scale and held by two small flanges. When I gently pulled the loose foot away I noticed that a small black wire had broke. I used my fingernails to remove a small bit of the wiring cover and twisted the ends back together. I then hide the wires back in a recessed area on the foot and placed it back on the scale. Worked just like the day I got it. Hope this helps!
Posted on Feb 20, 2009
a 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of.(from cars to computers, handyman, and even drones)
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Sep 12, 2016 | Conair Weight Watchers WW-87 Body Fat...
Aug 26, 2013 | Health & Beauty
Nov 29, 2011 | Weight Watchers Glass Electronic Scale
Sep 18, 2011 | Home
Oct 24, 2009 | Conair WW81 Weight Watchers Body...
Oct 05, 2009 | Conair WW81 Weight Watchers Body...
Oct 04, 2009 | Conair Weight Watchers WW-87 Body Fat...
Mar 28, 2009 | Conair Weight Watchers WW-87 Body Fat...
Jan 28, 2009 | Conair Weight Watchers Glass Digital Body...
Feb 10, 2015 | Conair Weight Watchers Glass Digital Body...
2,443 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: