Question about Life Science Bionet Cardiocare 2000 12 Channel Ecg
My machine stopped printing the full EKG. It starts and then just stops when the paper is half out. I have tried replacing the paper roll, opening and closing the lid, etc. Any suggestions?
Hi, if i want to start running unit. after that always on display data error, plz turn off. why? what can i do for that? thax
Posted on Jun 30, 2013
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
SOURCE: hi,i have probleme in ekg bionet 2000, when i test it without electrods and start printing,,at first it print signal normal but after a few seconds he show a huge and large line with lot of noises in
We are experts in fixing the Bionet 2000 ekg machines.
1) Is the machine plugged in to the wall when you are getting interference? If so, unplug it
2) Your patient cables may be malfunctioning, do they look damaged?
3) Are the electrodes expired?
4) make sure and prep the patient
If you are still having issues, send the machine to us at Discount Cardiology. We will look at it for free and let you know what is wrong with it. You then would have the option of fixing the machine or having it sent back to you.
Posted on Oct 08, 2012
Tips for a great answer:
Jul 28, 2016 | Schiller Cardiovit At 1 Ecg Machine
Dec 21, 2015 | Medical
Mar 07, 2013 | Life Science Bionet Cardiocare 2000 12...
Apr 27, 2012 | Schiller Welch Allyn Cardiovit At 2...
Jan 30, 2012 | Life Science Bionet Cardiocare 2000 12...
Aug 09, 2011 | Ge Medical Systems Ge Marquette Mac 1200...
Aug 05, 2011 | Life Science Bionet Cardiocare 2000 12...
Jul 06, 2011 | Schiller Cardiovit At 1 Ecg Machine
569 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: