Question about Seca 700KG Balance Beam Scale With Wheels, 220 Kg Max
Do we need to be greasing or oiling something. When a person of 300 lbs. gets on the scale, something feels like it is rubbing and doesn't want to give a true weight. Also I can't seem to find a manual for the seca 700 on the website. Am I crazy? Thanks
Checked the bottom of the scale to see if all loops are in place in the four corners. if they are in place load the scale with a body weight the heavier will be better and the look in between the platform and the frame see if there is room. if the platform is rubbing the frame the scale is worn out. chipper to buy a new one.
Posted on Jul 28, 2012
I contacted the manufacturer and here is there reply.
No need for oiling or grease, but tip the scale on its side, and look
under the base. There will be four "connection points" where pivot pins
connect to the linkage assembly; see if any of those pins or linkage
points are bent or look different from the other ones.
Then, note down the serial number of the scale located on the side of
the base, or behind the column of the scale at the top or bottom of the
column, and call our Maryland repair center at 800-542-7322 ext. 111 and
speak with Tyson Long from 9:00 AM - 4:00 PM Maryland time to describe
the issue and for further troubleshooting over the phone.
I have attached our new catalogue in PDF format as well. FYI: The model
703 is a very popular, extremely heavy-duty, high-capacity digital
column scale that is a popular alternative to the mechanical physician
scales in the gym and schools environment.
seca north america west
Medical Scales and Measuring Systems seca corp.
3401 Centrelake Drive, Suite 320
Ontario, CA 91761
Telephone +1 909 605 7600
Telefax +1 909 605 7654
Posted on Mar 16, 2009
Tips for a great answer:
Jun 07, 2013 | Health-O-Meter Health o Meter BFM681DQ-63...
Sep 13, 2011 | Health-O-Meter Health o meter 400EXP...
Mar 11, 2011 | Seca 710 Waist High Mechanical Balance...
Jan 30, 2011 | JR Ohaus -300 Ruby Jewelry Scale, 300 x...
Dec 27, 2010 | Health-O-Meter Health o meter 402KLWH...
Dec 10, 2010 | Homedics Glass Digital Scale - Advanced...
Aug 08, 2010 | Health-O-Meter 500KL Medical Scales...
Jan 16, 2010 | Jobar International Idea Works Wireless...
Jul 14, 2009 | Soehnle 63700 Calando Quattrotronic Scale
Jan 08, 2011 | Seca 700KG Balance Beam Scale With Wheels,...
385 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: