Question about Invacare Health & Beauty
I'm helping a Veteran that the VA administration has left hanging. He has an InvaCare power chair TDXSP. The controller is showing a "left brake fault". I unplugged both motors from the power control box on the bottom back of the chair and reversed them. Then it said I had a "right brake fault" when I powered it up. I see a couple of solutions here about replacing the left motor assembly under this same question. Would this still be a solution for this problem or is it the motor control box that they are plugged into need to be replaced?
Hi, Please check the lead and the fitting to the controller, it 's rare for the controller to have this problem, I have repaired that very fault recently and it was a high resistance cable fault , now measure the brake cable from the controller , disconnect cable and put the electrical meter on the lowest ohms reading, now it should read from 0.4 to 1.0 ohms any more and that's what causing the problem.
Posted on Jan 20, 2013
a 6ya Expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repair professionals here in the US.
click here to Talk to an Expert (only for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
SOURCE: no forward or reverse. lights
Joe with a scooter there can be a number of things wrong to cause your problem Since the motor turns and you have power at the console. It could be the throttle pot or the speed pot or both or it could be the cable that runs from the console to the rear of the scooter and finally a problem with the controller. But that would also give some sort of error to the console light if it is equiped with a power light. If Medicare paid for your scooter then they will pay most of the cost of repair. Your provider can work on it for you or they can contact a repair service that will fix it for you Hope this helps
Posted on Sep 27, 2010
SOURCE: We replaced brushes on motors
GOD I LOVE BEING HERE FOR PEOPLE LIKE YOU,PLEASE DOWNLOAD :SHOPRIDER SERVICE MANUAL 2007 1-800-743-0772 THAT WILL GIVE YOU WHAT YOU NEED,COLIN.
Posted on Mar 14, 2011
SOURCE: I have a Invacare M91
Why not just lock the joystick. you can have the provider check the programming and you should be able to put it in lock mode then unlock it when you want to use it Hope this helps
Posted on Apr 20, 2011
SOURCE: gentleman was using the Storm
Hi Please check this link for the fault code:- Please get back to us if you have further query else please accept the suggestion.
Thank you for contacting fixya.com
Please check this link for the fault code:-
Please get back to us if you have further query else please accept the suggestion.
Posted on May 03, 2011
I'm in a power wheelchair myself. If the wire coming out of your controller has a place in which it can be unplugged, you can try unplugging that, then plugging it back in. If that doesn't work, then you probably don't have the appropriate tools needed to reset the controller. You'd need to take it in to get serviced.
Posted on May 03, 2011
Tips for a great answer:
Apr 03, 2016 | Electric Mobility Invacare Pronto M41...
Jul 15, 2014 | Electric Mobility Invacare Pronto M41...
Dec 27, 2013 | Jet 3 Ultra Power Chair
Oct 28, 2012 | Health & Beauty
May 01, 2012 | Jet 3 Ultra Power Chair
Sep 16, 2011 | Invacare Supply Group Pronto M61 Standard...
Jun 25, 2011 | Invacare Pronto M51 Power Wheelchair
Mar 17, 2011 | Invacare Pronto M51 Power Wheelchair
Mar 06, 2011 | Invacare Health & Beauty
Jan 21, 2018 | Invacare Health & Beauty
288 people viewed this question
Usually answered in minutes!