Question about Health & Beauty
a 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of.(from cars to computers, handyman, and even drones)
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
SOURCE: RESPULES Question
No, PULMICORT RESPULES should only be delivered through jet nebulizers. Jet nebulizers are the only effective device for delivery of suspensions such as PULMICORT RESPULES. Because of their physical characteristics, suspended drug particles are generally poorly nebulized in ultrasonic devices.
Posted on Aug 27, 2008
SOURCE: nebulizer filters
The nebulizer you have is made by Invacare Supply Group. You can order online at onlinemedicalsupply.com or you can call them at 1-800 860-0305. Have a nice day. and please dont forget to rate me.
Posted on Sep 30, 2008
The long tubing should have a green end and a blue end. (If both ends are the same it does'nt matter) Now the little cup that holds the medication...look at the bottom of the cup. It should have a male connector (blue or green) take the end of the tubing that matches the color of the male connection and firmly plug them together. Hold the chamber/cup upright. If your unit has a little plastic diffusor, drop that inside the chamber. now locate the top af the chamber and screw it on. You should have a large whole the size of a nickle on top. This is where you squeeze the medication in. After putting in the medication put the mask(or mouthpeice) Firmly into the place where you put the medication into....and your peices should be assembled. Last thing is to plug the other end of the tubing into the machine itself and then turn on the power.
Posted on Dec 23, 2008
SOURCE: my devilbiss nebulizer is not
Here is a link that has some trouble shooting ideas...I hope this helps.
Posted on Jan 07, 2009
You'll read a lot about making sure that the filter is clean and that the medicine cup is cleaned and that hard water can effect the operation of the nebulizer. However, the problem is most likely in the pump. Even though the pump is running, it is not pushing enough air into the medicine cup. You may even feel air coming out of the tube. When unrestricted, the air will come out, but, once restricted by the medicine cup, the air may not be fast enough to nebulize the liquid. If you want to confirm this, take the tube and hold it to the end of a bike pump, if, while pumpung, you successfully see mist, then you know for sure.
Fixing the pump is not too difficult. Over time, the chamber where the air is compressed becomes dry and, just like a dry bike pump, the air escapes from the chamber due to poor seals. The pump is pretty simple. Make sure it is unplugged from power and remove the screws holding the housing together. Inside you will see the motor, a fan, a power switch and the intake and outtake tubes. Follow the outtake tube to the pump. The junction where the tubes meet the pump is the location of the air chamber. You may need to remove the screws that secure the tubes to the chamber. Once apart, take some grease and lightly coat all of the interior surfaces of the chamber. When finished, reassemble and you're good to go!
Posted on Apr 21, 2009
Tips for a great answer:
Feb 16, 2013 | Medline Sport Mist Compressor Nebulizer...
Aug 06, 2012 | Omron Health & Beauty
Aug 06, 2012 | Omron Ne U22V Microair Vibrating Mesh...
May 17, 2010 | Mabis 40 270 000 Mist II Ultrasonic...
Jun 25, 2009 | Omron Healthcare 1 OF Compair Elite...
Apr 20, 2009 | Pari Proneb Ultra II Compressor Nebulizer...
Apr 07, 2009 | Mabis Neb Xp Portable Handheld Cmpressor...
Aug 27, 2008 | Cardinal Airlife Brand Misty Max 10...
141 people viewed this question
Usually answered in minutes!