Question about Celestron Advanced C9 1/4-S 11045 (555 x 235mm) Telescope
When I attach any component ie; adapter, diagonal, extender etc to the visual back, the whole tube rotates with the equipment. This causes the eyepiece and all to flop down while observing at an angle. It is evident that the baffle tube / visual back base has worked it's way loose. How can I re-tighten this? If dismantling the scope is required is there a locking nut or grub screw responsible? Any recommendations would be helpful.
Hi David, the visual back should thread on to the rear cell of the OTA but on the visual back there should be a screw to lock in you 1.25" accessories.
Posted on Nov 05, 2009
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Sep 07, 2014 | Optics
Aug 19, 2011 | Optics
Jul 25, 2011 | Celestron PowerSeeker 60 Astronomical...
Mar 25, 2011 | Celestron NexStar 6 SE (354 x 55.88mm)...
Sep 29, 2010 | Celestron Ultima 2000 -- 8 Telescope
Jan 28, 2010 | Lomo Astele 150 OTA (300 x 150mm)...
Oct 13, 2009 | Meade LX50 (8105003) (500 x 203mm)...
Sep 20, 2009 | Celestron Optics
Jun 22, 2008 | Celestron NexStar 4 SE Telescope
162 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: