Question about Sekonic L-358 Flash Master Light Meter
When I start my sekonic on the computer an error message on the screen says Screen number Frm Main Object name fGrtlnispring() Subfunction name setmenu caption() Error nimber 339
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
I've known this problem to occur if the wrong kind of CR123/CR123A lithium battery is used. They come as rechargeable and dry cell (i.e. non-rechargeable) versions. For sensitive devices like yours the difference is important as the internal resistance of the rechargeables is lower which affects the voltage characteristics.This meter is designed only for the dry cell version: see page 34 of the user manual, although the instructions don't make this fact clear enough IMHO. The reason for using the dry cell version is that the output voltage remains constant until the battery is nearly exhausted, but the rechargeable version starts at a slightly higher than nominally rated voltage and drifts down as the battery discharges. This means that you only get a truly accurate reading with a dry cell battery. There is no user adjustment to recalibrate the meter but you can use a different ISO rating to compensate or use the exposure compensation settings.
Posted on Jul 04, 2009
have you checked your sync cable?
tried meter on another system (for comparison)?
Depends where you are.. Mamiya in the USA can solve the problem.
Posted on Jul 21, 2009
This is the link to the PDF manual: http://www.sekonic.com/upload/files/L-758_Operating%20Manual_English.pdf
Cheers, hope that helps!
Posted on Jul 03, 2010
SOURCE: Re: Sekonic L-328. Sometimes the
You could be having a switch problem, or another intermittent problem. Most the the switches on devices these days require that the unit be disassembled to clean them, and that's tedious. You should call Sekonic and see if they can help, or service the unit.
Good luck, and hope this helps.
Posted on Apr 21, 2011
Tips for a great answer:
Apr 01, 2014 | Sekonic L-308s Flashmate Light Meter...
Aug 29, 2010 | Sekonic L-358 Flash Master Light Meter
Mar 07, 2010 | Sekonic L-358 Flash Master Light Meter
Oct 12, 2009 | Sekonic L-358 Flash Master Light Meter
Jun 08, 2009 | Sekonic L-358 Flash Master Light Meter
Apr 19, 2009 | Sekonic L-358 Flash Master Light Meter
Mar 06, 2009 | Cameras
Dec 20, 2007 | Casio Exilim EX-S500 Digital Camera
40 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: