Question about Schiller Welch Allyn Cardiovit At 2 Interpretive Ekg Machine
I try to do a ekg on a patient and the machine will go off please explain what to do thank you
Or dirty pole connet cord shall cause you chang expensive batt,try take it out for new outlet charger and inspect said point !
Posted on May 27, 2013
Greetings. there is an internal fuse that is blown, and you battery is depleted. The AT2 will shutdown when the paper drive starts when the battery is weak enough. Your unit needs a battery and a PC board repair. The fuse is not operator changeable, nor is the battery. . Unfortunately, there is nothing you can do with one exception: Try a new lint cord, and make certain that the green AC LED is illuminated when the unit is plugged in. If so, try keeping the unit plugged in for a few hours. If proper operation does not come back, you have internal problems that require a skilled electronic/biomedical tech to re-mediate. In that case you should seek professional assistance in the dis-assembly and servicing of your unit.
Posted on Nov 21, 2011
There is an internal battery in this machine. Machine reads that first. You need to replace that battery. It is almost $30.
Posted on Jan 23, 2009
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Nov 15, 2011 | Ge Medical Systems Ge Marquette Mac 1200...
Nov 03, 2011 | Schiller Welch Allyn Cardiovit At 2...
Mar 31, 2011 | Schiller Welch Allyn Cardiovit At 2...
Feb 14, 2011 | Schiller Welch Allyn Cardiovit At 2...
Feb 04, 2011 | Schiller Welch Allyn Cardiovit At 2...
Jan 24, 2011 | Medical
Jan 10, 2011 | Life Science Bionet Cardiotouch 3000 12...
Jul 18, 2010 | Schiller Welch Allyn Cardiovit At 2...
Apr 05, 2009 | Schiller Welch Allyn Cardiovit At 2...
517 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: