Question about Conair Weight Watchers WW-87 Body Fat Glass Scale

# BF% is wrong

The scale is on a flat level tile floor.  I step on it without clothes on, and barefoot.  All of the user info is entered correctly.  I am a male, 23 years old, 5'9 and 150 lbs.  I barely have a inch of fat on me.  It is telling me that I have 18.9% bf.  I had my bf% measured at GNC, and their scale read 8.4%.  Is this just a faulty scale?

Posted by on

• Level 1:

An expert who has achieved level 1.

Mayor:

An expert whose answer got voted for 2 times.

• Contributor

Unfortunately, it's the scale, but it's not broken. It simply is telling you inaccurate information.
Some scales utilize Bioelectrical Impedance Analysis (BIA) technology. BIA uses very low levels of electrical impulses running from one metal contact point/lead to another. Your body acts as the connector between the two contact points, thus completing the circuit. Not all scales use this means for calculating body fat percentage due to poor reliability and high fabrication costs.This method of measurement requires very large contact points, as well as clean and moist skin exposure. If you have dry/calluced feet, which is normal for most Americans, the sensors can not accurately measure due to skin resistance. Dry skin acts as a resistor to electrical impulse. Electricity passes through muscle and fat at different rates. However, its measurements may be thrown off by how hydrated you are and how much you've eaten; it may be too inaccurate for anything other than comparative use. Users are encouraged to establish a baseline measurement and repeat measurements under consistent conditions over time to detect changes.
Many less expensive models simply use a body-fat calculator to average your imputed height in inches, waist measurement in inches and the current weight measurement recorded in lbs. The scale compares your measurement with the ideal norms inputed during the manufactoring process, thus giving you a comparison vs.imputed norms. People who are very, very fit can get numbers 3 to 5 percent higher than their true percent body fat. Because they don't have a lot of fat inside their muscles(visceral fat), the simple tape measure calculation test cannot adjust as it should indicate.
Conversely, if a person is skinny but not fit, this body fat test may yield a number 3 to 5 percent lower than his or her true percent body fat. Though they look thin, unfit skinny people really have more than the usual amount of fat inside their muscles, which you can't see from the outside.
In either case, the scale is not necessary a measurement tool, due to high variables of test inaccuracy and interuser reliabilty; but, it is a good tool for comparing whether or not you are gaining/losing body fat.
If you would like a more accurate tool for simply measuring your body fat, you can pick up some pinch-calipers online for about the price of a scale. They come with an instructional pamphlet, very easy to use.

Posted on Jan 13, 2009

Hi,
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
goodluck!

Posted on Jan 02, 2017

×

my-video-file.mp4

×

## Related Questions:

### I have Hanson HX5000 bathroom scales that are showing error message OL. I have replaced the battery and ales no difference - anyone gave any ideas how to fix? Photo for reference

First, make sure the battery is not in backwards. If that's no help, let it set overnight without the battery in it before putting it back in. Also, make sure the unit is setting on a level, solid floor - not carpet (this looks like it's tile, but the grout lines are awfully bumpy-looking. . . If none of these things work, I'd say you got a bad unit. Were there no instructions included?

May 14, 2016 | Health & Beauty

### I have two Taylor Lithium scales and neither one of them will turn on. I have replaced the batteries, still they will not turn on. I have done a "reboot" as suggested on FixYa, but without...

The scale may need to be initialized. This process is necessary because the scale was moved during the shipping process. Initializing the scale "re-seats" the internal parts allowing the scale to find the correct "zero." Here is how to initialize your scale:
- Place the scale back on a hard surface floor.
- With one foot only, place just enough weight on the scale for the display shows zero's or dashes. Once you see zeros or dashes remove your foot.
- Wait for the scale to turn off again.
- Step up onto the scale for a weight.

Or, your scale may be on an uneven surface or on carpet. Be sure to place your scale on a hard, flat surface to get an accurate weighing. Are you sure that the place the new battery with the "+" sign up into the battery compartment?

Keep us updated.

Jul 15, 2011 | Taylor 5599 440 Pound Tempered Glass Body...

### I have a Health O Meter HDM560. I need to reset the #1 and #2 users and start over, can you help me there is nothing in the manual about it I can find.

Current Instruction Manual for the HDM560 can be found here:

http://www.healthometer.com/Manuals/MANUALS/HDM560-05_43_41816518.PDF

For resetting it would be similar to the initial setting process. FOLLOW these steps:

Lift display cover. Push "SET" button at bottom of pad. User "1" and "2" will appear.
User "1" will start blinking. Press the "USER" button to choose user "1" or "2". Once selected, press "SET" button to lock in information.
Both "HEIGHT" and "AGE" will appear on the screen. The "HEIGHT" will start blinking. Press the "UP" or "DOWN" arrows to find the correct height.

After entering height, press the "AGE" button. Use the "UP" and "DOWN" arrows to enter correct age and press "SET".

Both "MALE" and "FEMALE" images will be displayed. Press the "GENDER" button until the correct MALE/FEMALE image blinks and then press "SET". This will lock in the information.
Your correct information including User, Height, Age, and Gender will be displayed. Now you are ready to calculate weight.

Tap your person "USER" number (1 or 2) in center of scale with your toe. Your personal profile will appear. Wait until "000" appears. Stand on scale so bare feet are positioned over silver pad surface. "000" will blink your weight for approximately 3 seconds. "BF" will flash and body fat percentage will appear. The scale will turn off automatically in a few seconds.

May 21, 2011 | Health-O-Meter Health & Beauty

### The weight jumps around. If I get off the scale and on again, the weight changes by 5 pounds. I am sure neither weight is correct.

You're only suppose to weigh yourself once a week. Make sure you're doing it after you've gone to the bathroom in the morning and with no clothes on. Other than that, I'd make sure the floor you set the scale on is level and that you are holding very still while weighing.

Feb 28, 2010 | Conair WW60 Weightwatchers Scale

### I just purchased a Healthometer Weight & Body Fat Monitoring Scale. I successfully setting up my scale, I think? When I turn the scale on and weight for the 0.0 to appear, I get on the scale and it...

These scales use sensors to measure the electrical resistance of the human body to calculate body fat. If you are not stepping on the scale barefoot then that might be the reason why it is timing out. These scales require you to step on it barefoot so it can take a measurement through your feet. And using the height and gender information in combonation with the resistance measurement it can calculate your approximate body fat precentage. If you are stepping on the scale barefoot then make sure that your feet are properly placed into the sensors and avoid any movement. Let me know if you have any questions.

Feb 06, 2010 | Health & Beauty

### Scale is not weighing correctly. Weighs correct

The scale must be on a hard, level surface. Make sure there is absolutely no dirt on any of the pivot points. Calibrate it with a KNOWN 25 pound weight.

Jan 04, 2010 | Health-O-Meter Health o meter 402KLWH...

be sure you are barefooted when weighing yourself

Oct 05, 2009 | Tanita BF-350A - Body Composition...

### Err 4

Sounds like it wasn't initialized, even though it says Er 4 instead of Er 0... According to my manual, to (re)initialize it and reset it to zero, tap the platform lightly. It should show zeroes running across the display for a second or two, then show all zeroes and turn off. It should be re-initialized...

Error Codes for the 5757 (according to my manual):

Er 0 Initialization error (remedy described above)

Er 1 Instability error (step off the scale, then back on)

Er 2 Overload warning (registering weight beyond the 440 pound max)

Er 3 Contact error (body impedance beyond technical limit - try dampening feet and try again for better contact)

Er 4 Bodyfat error (bodyfat percentage is beyond range - try dampening feet and try again for better contact)

Er 5 Bodywater error (bodywater percentage is beyond range - try dampening feet and try again for better contact)

Jul 02, 2009 | Taylor 5757 Digital Glass Body Fat Scale

### How to utilize the WW87 Glass Body Analysis Scale?

I discovered that it helps if your feet are slightly damp. I usually wipe them with a washcloth before getting on the scale. They don't have to be WET, just every so slightly damp.

Dec 20, 2008 | Conair Weight Watchers WW-87 Body Fat...

### Tile floor installing

you need to have all the equipment and goods. The tiles, cement to fix them, the cutter, a level, a rubber hammer, a marshy, a water drum, and paper.
1. use the paper to cover all the area that may be damage.
2. level the surface
3. put the title in the level floor, without pasting them, and measure the titles you need to cut, to fix all the are, measure and align them.
4.with the marshy apply the cement to the area
5. Put the titles and with the robber hammer hit them softly.
6. went finish clean them with a soft rag and water.

Oct 02, 2008 | Measuring Tools & Sensors

## Open Questions:

#### Related Topics:

558 people viewed this question

Level 3 Expert

Level 1 Expert