Question about Fluke 705 Process Calibrator,current And Voltage
Hello. I have a blown fuse and relay on my fluke 705. Does someone can tell me the reference of this fuse and relay? F1 and U11 on the pcb. Thank you. http://img687.imageshack.us/img687/6969/26102012893.jpg http://img59.imageshack.us/img59/7417/26102012894.jpg
F1 (Fluke PN:800901 Ver. A,B and C) (PN: 1778294 Ver. D & E) is 350mA 600V fuse.
U11 (Fluke PN:687555) is DPDT Latching 5 Volt Coil
I would also check D12 if your revision has one (PN:1617521) which is sidactor and it's other side of the board (DO-214 58V).
Posted on Oct 29, 2012
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Hi, that is the error message indicating a EEPROM error and unfortunatlely, you can't do anything except have the meter serviced by either Fluke directly, or by an authorized Fluke test equipment dealer.
Since Fluke doesn't provide the specs and coding for their products, you wouldn't be able to repair the meter properly, even if you could burn and program your own chipset.
Here's a copy of the manual for your meter, so you can contact Fluke for servicing and see the other info relating to it (for your future reference)
Sorry to be the bearer of bad news, but Fluke products are typically quality test instruments and there might be a possibility that you're still covered under their warranty, so checking with them is the best avenue for getting it back to working condition.
I hope you find this Very Helpful and best regards!
Posted on Nov 10, 2009
Error 1 is an indication that the self test has failed. This will require repair at an authorized Fluke service center. There is an internal problem with the meter. The only thing that you can try is to replace the internal battery. Should the battery be defective, the internal testing can be effected. I have only seen this occur once, but it is worth the attempt. Should a new battery not fix the problem, return the meter to Fluke for repair.
Posted on Apr 15, 2010
Go take a look here please: http://www.fluke.com/fluke/usen/support/default.htm I found your manual here: http://www.fluke.com/Fluke/usen/support/manuals/default.htme: but you may not be able to go to this directly. Click on the Manuals tab then search your model number.
I trust this is what you are looking for.
Thanks for your interest in FixYa.com
Posted on Feb 14, 2011
Tips for a great answer:
Jan 30, 2013 | Fluke Networks Aircheck Wifi Tester
Jan 23, 2013 | Fluke 725 US Multifunction Process...
Jan 15, 2013 | Fluke 744 Documenting Multifunction...
Dec 16, 2012 | Measuring Tools & Sensors
Jun 07, 2012 | Fluke 51-2 Thermometer Series II
Mar 04, 2011 | Fluke Networks Fluke T5 600 Elec. Tester...
Dec 13, 2010 | Fluke Networks Fluke Fluke 789
Nov 01, 2009 | Fluke Networks Fluke 179 Electricians...
May 20, 2009 | Fluke Networks Fluke T+PRO Electrical...
538 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: