Question about Fisher Science & Laboratory
It will only read 20.00 or -2.00 on the pH setting. Occasionally it will read other values that appear correct, but only for a split second before jumping wildly and then resolving to -2 or 20. Following procedures to run calibration are unsuccessful. Using electrode 13-620-531. Will take temperature accurately. I would hate to track down a new electrode if there is a simple solution I'm overlooking. (I tested the meter by capping the electrode input and checking the voltage, which was accurately 0). Any help would be greatly appreciated.
The electrode you have is a combination electrode with an integral calomel reference half-cell and built-in temperature probe. Since the temperature reading is good, the temperature probe is okay. It isn't part of the pH circuits anyway. (Your meter uses the temperature to calculate a temperature correction of pH.) Something else is going on with the electrode proper. Its funny behavior suggests to me that it has dried out internally.
pH electrodes don't last forever. How old is this one? If it's an old one, it's probably time to replace it. If it's new and doesn't work, it may be under warranty.
If you shop for a new electrode, get a gel-filled one. And go for one with a silver/silver chloride reference instead of a mercury one -- especially if you are using it on food or other samples that may be consumed. Another consideration about calomel electrodes: they take hours to recover from relatively small temperature changes. Ag/AgCl references aren't as sluggish.
Finally, take care when storing the electrode. It should be stored under humid conditions to keep it from drying out. The best way is to put a small piece of cotton or glass wool soaked with distilled or deionized water inside the plastic end cap. For shorter-term storage, e.g., overnight, you could immerse the tip in pH 7 buffer.
Posted on Nov 22, 2017
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
ok my friend, right-click on the desktop and select the properties tab then go to display settings/settings and adjust your resolution to 1024x768 click apply and save, you should now have a decent display
Posted on Nov 15, 2008
Take your temp knob off and look in the center of the stem and you should find a small set screw. Using a small precision screwdriver, turn the screw until you see the light come on. (You won't have to turn it far.) Turn it a little at a time, maybe a 1/16 of a turn. Leave your thermometer in the oven with the door closed and at or near the temp probe that is in the oven. This way, you are getting a true reading.
Posted on Feb 15, 2009
Tips for a great answer:
Oct 09, 2017 | LW Scientific Urine Reagent Strips For:...
Aug 08, 2016 | Fisher Scientific 15-077-8 - Scientific...
I appreciate your post, Ronald!
From the information you have provided you would like to calibrate the oven temperature on your Frigidaire range. I was unable to locate any instructions on how to perform a calibration on your oven. I would recommend consulting with a professional technician for proper calibration of your range. I hope this information is helpful. -Matt
Jan 01, 2015 | Frigidaire Kitchen Ranges
Dec 05, 2014 | Measuring Tools & Sensors
Jun 27, 2014 | Oregon Scientific WMR86A Backyard Pro...
Mar 18, 2013 | Casio fx-115ES Plus Scientific Calculator
Jul 27, 2017 | Lands Phones
Feb 28, 2011 | Hanna pHep pH Meter with Solutions
Aug 21, 2009 | Audio Players & Recorders
Mar 01, 2008 | Blaupunkt DX-V GPS Receiver
1,024 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: