Question about EKG/ECG Machines
Power is there on indication.
Tips for a great answer:
Nov 25, 2014 | EKG/ECG Machines
Power the Spot Vital Signs
off, wait five seconds, and power on. If the error code
reappears then power the
Spot Vital Signs off and disconnect the battery for five minutes.
Reconnect the battery and
power on. Press the Blood Pressure Start/Stop button to reset
flashing patient alarm conditions.
Dec 29, 2010 | Welch Allyn Spot Vital Signs Monitor with...
Jun 12, 2010 | Welch Allyn Spot Vital Signs Monitor with...
Jan 14, 2010 | EKG/ECG Machines
Feb 20, 2009 | Welch Allyn Atlas Monitor W/Ecg, Nellcor...
Feb 17, 2009 | Haba Schiller AT1 Interpretive Ekg, Each
Jan 29, 2009 | Welch Allyn Spot Vital Signs Monitor with...
212 people viewed this question
Usually answered in minutes!