Question about Starrett Measuring Tools & Sensors
How do you check a starrett 98 level to ensure it is adjusted correctly?
You do not need to replace it. If it does not read the same when you reverse it, on one end there are a nut over and under the tube and plug assembly. Adjust those up or down until it reads the same.
Starrett Tech Support
Posted on May 15, 2009
Hi, Easy fix. Put the level on any surface, note where the bubble is, Then turn the level 180 degrees and check the bubble again. If it's in the same place, Your level is perfect. If it's not, unfortunately you will need to replace it.
If this doesn’t work for you, please write back.
Posted on Nov 29, 2008
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Nov 12, 2010 | Starrett 50443 12 Machinists' Improved...
Dec 27, 2009 | Starrett Measuring Tools & Sensors
Dec 07, 2009 | Starrett Electronic Caliper
Oct 15, 2009 | Starrett 50443 12 Machinists' Improved...
Aug 06, 2009 | Starrett 50443 12 Machinists' Improved...
May 04, 2009 | Starrett 50443 12 Machinists' Improved...
Mar 07, 2009 | Starrett T216XRL-1 - MICROMETER, DIGITAL
480 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: