Question about Wesco WS7 Microscope
It looks like when you took this apart you did not alignt the lenses properly, there should be a locating groove or stub which these lenses should fit to... it looks like they are pressing against the stub but are not located properly, hence the distortion. try taking it apart again....when you put it back together again the fault may have gone.
Posted on Sep 15, 2007
Tips for a great answer:
Aug 02, 2014 | Tasco 750 Monocular Microscope
Feb 05, 2014 | Bausch Lomb Bausch & Lomb Stereozoom 7...
Nov 12, 2010 | Ken A Vision T2600C Microscope
Oct 05, 2010 | Premiere Microscopes P-01310 Monocular...
Dec 22, 2009 | Tasco Microscopes
Nov 28, 2009 | Wesco Microscopes
Dec 15, 2008 | Swift Microscopes
261 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: