Question about Invacare Pronto M51 Power Wheelchair
Posted by Anonymous on
a 6ya expert can help you resolve that issue over the phone in a minute or two.
best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
the service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
SOURCE: oil / gel leak
If it is coming from your motors, it could be very bad. i.e. motor replacement. Sometimes on the side of those motors there is a black plastic cap about the size of a quarter or half dollar, those fall out sometimes and allow gearbox grease to come out. This should`nt be confused with your motor brush caps which are about the size of a pea and have a small **** in them for a screwdriver.DO NOT TAKE THESE OUT. If the large cap has come out just stick it back in. If you can`t find it you`ll ned to call a dealer to get the part replaced. If you can still find no prob,your probably going to have to go to a dealer.
Posted on Mar 26, 2009
SOURCE: won't move pronto m51
Three lights flashing on your chair indicate a left motor fault. Make sure that all of the connections inside the shroud are tight by physically testing them. If all of the connections are good, and the fuses are good, then it's time for a trip to an authorized repair facility. Here is a link to the M51 owners manual if you haven't one. Page 24 has the fault code guide.
The controller monitors all of the functions on the chair, brakes, motors, joystick, charger etc. The three flash fault indicates an "open" circuit. Open, like a switch or blown fuse is open, or a wire is cut or burned in half, or a connector is unconnected, or partially connected. The latter is what you are hoping for. Wires burned in half is the usual cause for a motor fault, and with powerchairs, is usually the result of the chair getting stuck in or against something it can't get out of and the operator over-amping the motors. Motors do, just go bad sometimes, for no reason. Very rarerly, however.
You can also check the brushes on the motor too. There's two quarter-sized black plastic screw-plugs on the ends of the motor opposite each other. Unscrew them and examine the brushes. The springy coils should be springy. Overheating can cause them to be brittle, or even burned in two. Check the lench of the brush, the dark rectangular cube at the end of the springs. It should be curved and polished at the end, and a little over a half-inch long. If it is near a 1/4 inch, it needs replaced, or if there is any evidence of overheating, replace. Good luck.
Posted on Jul 09, 2009
SOURCE: My wheelchair dose not move
The chair has two levers, one for each motor, that disengage it for freewheeling, or pushing the chair without power. These take it out of gear. Place the levers in the locked, gear-engaged position and it should go for you. Here's a link for the manual for your chair, and it will show you where the location is to place it in gear.http://www.invacare.com/doc_files/1125085J.pdf
Posted on Jul 12, 2009
SOURCE: speed control knob has come off
you need Invacare part # 1040225, cost about $6.00. Call these people 866-722-4581
Posted on Sep 22, 2009
Testimonial: "I looked on the internet for this part for HOURS without success. The "FixYa" expert found the model number needed and the phone # to order. Thanks!! "
Tips for a great answer:
Dec 25, 2015 | Health & Beauty
Feb 12, 2014 | Invacare Pronto M51 Power Wheelchair
Dec 15, 2013 | Invacare Pronto M51 Power Wheelchair
Aug 11, 2010 | Invacare Supply Group Invacare Pronto M51...
Jul 11, 2010 | Invacare Pronto M51 Power Wheelchair
Oct 17, 2009 | Invacare Pronto M51 Power Wheelchair
Apr 21, 2009 | Invacare Supply Group Invacare Pronto M51...
Apr 16, 2009 | Invacare Pronto M51 Power Wheelchair
Apr 16, 2009 | Invacare Supply Group Invacare Pronto M51...
67 people viewed this question
Usually answered in minutes!