Question about Mitutoyo 0-18"mitutoyo Electrnc Digi-matic Height Gage
I can't get my Fluke meter 8060AA to display anything. I worked when I last used it 8 yrs ago. I replaced the 9v battery with a new good battery and cleaned the contacts. At times I could see a very dim display. Now nothing displays when it's turned on. Any ideas??
Test unit reads okay then goes into - OL mode unable to take any readings
Posted on Jul 12, 2011
The fuse has nothing to do with the display.
The contacts of the LCD are probably oxidized because of its age. With the unit out of its care, carefully remove the LCD from its retaining mount. Use rubbing alcohol and a clean q-tip to clean the contacts of the LCD and the small "rubber-like" strips that contact both sides of the LCD. Also, remove the strips and clean the other side and the contact on the main circuit board. Then reassemble.
*Note, if you live in a cold climate, freezing of the LCD could also cause this problem. In that case you must purchase a replacement LCD from Fluke.
Posted on Jan 25, 2009
Open it up .there is a fuse n the inside
Posted on Oct 12, 2008
Tips for a great answer:
Jan 24, 2013 | Fluke Networks Fluke 87-5 Digital...
Nov 11, 2011 | Fluke Networks Fluke 87-5 Digital...
Jan 06, 2010 | Fluke Networks Fluke 336A Digital Clamp On...
Oct 26, 2009 | Fluke Networks Fluke 336A Digital Clamp On...
Nov 13, 2013 | Mitutoyo 0-18"mitutoyo Electrnc Digi-matic...
253 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: