Question about Fedders A7DH65B2A Dehumidifier
Posted by Anonymous on
You can test the humidistat fairly easily, using a volt-ohm meter. Just set the meter to RX1 scale and attach its leads to the humidistat's terminals. Rotate the humidistat's knob as far as it will turn in both directions. If the volt-ohm meter registers zero ohms through only part of the humidistat dial's range, the humidistat is working. If it registers zero through the entire range, it's broken and must be replaced. Dehumidifier Overflow Cutoff Switch Test- Using a volt-ohm meter, you can tell if the overflow switch works or not. Clip the leads to the terminals on the switch. Depress the bar or trip lever on the switch. If the meter's needle doesn't show continuity and no continuity as the switch is clicked back and forth, the switch is probably faulty and will need replacement.
Posted on Jan 18, 2006
A 6ya expert can help you resolve that issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
The service is completely free and covers almost anything you can think of (from cars to computers, handyman, and even drones).
click here to download the app (for users in the US for now) and get all the help you need.
Posted on Jan 02, 2017
Tips for a great answer:
Oct 30, 2015 | Dehumidifiers
on Sep 15, 2010 | Dehumidifiers
Aug 19, 2010 | Maytag M7DH65B2A Dehumidifier
Sep 07, 2008 | Comfort-Aire BHD-501 Dehumidifier
Jun 29, 2008 | Danby DDR586R Dehumidifier - 58 Pints...
Nov 10, 2007 | DeLonghi DE400P Dehumidifier
Sep 06, 2007 | Kenmore 54501 Dehumidifier
Jan 18, 2006 | Fedders A7DH65B2A Dehumidifier
Jun 14, 2013 | Fedders A7DH65B2A Dehumidifier
745 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: