My machine will not turn on
Greetings. The schiller AT2 is battery powered. It will power on when plugged in however even with a dead battery, but if the battery is REALLY bad, we have seen it load the power supply preventing power up. On the top of you unit there is a small green "AC" indicator that should be illuminated when the at2 is plugged in. If it doesn't, your power cord might be bad, try a new one. If that doesn't fix your problem I would try at least to replace the battery. The battery requires dis assembly to replace, and there are three cable attaching the top case to the PC board. If you pull it apart to far without detaching the cables, you will do expensive damage. Also the removal of top case exposes high voltage, so this should never be undertaken a. with the unit plugged into a wall outlet, or b. By an unskilled individual. I would recommend that you find a local bio-medical contractor, or other service company, and have the battery changed. If your chosen contractor replaces the battery and the problem remains, your main board likely needs replacement, again by a qualified individual. Be warned however this in a very expensive part. There are several other possible problems, but very few that are operator serviceable. Best of luck!
on Nov 21, 2011